We were called into the office because the handpiece holder was leaking air and water.
Disassembly and Repair
After opening the delivery unit, it was found that the handpiece holder valve was malfunctioning and needed replacement. Adec provided an extra valve with the initial sale of this unit which we installed same-day into the handpiece holder.
Full Testing of Dental Unit
After installation of dental handpiece holder, we assessed all hosing, water lines, optics as preventative maintenance.
Bad Power Optic Tubing
Disassembled broken power optic dental handpiece tubing in DCI unit.
New Optic Hosing
New power optic dental handpiece tubing installed into DCI dental unit and fully functional.
This compressor was housed in the back-office with pre-built custom framing and piping. We were tasked with removal and replacement of the compressor without interfering with the existing infrastructure.
Removal and Cleaning
The old compressor was carefully lifted over the existing piping and removed from the space to be responsibly disposed of.
New Compressor Installation
Working with Tech West, we had a custom compressor built without handles and extra under-side clearance to fit the dimensions of the space. We carefully installed pneumatic and electrical connectors to insure proper functionality of the new unit.
Midmark Apollo Compressor
Compressor was not shutting off and kept blowing the safety valve. It was diagnosed that there was a faulty relay in the pressure switch.
Disassembly Part 1
Image of the faulty pressure switch. The relay contacts are stuck in the closed position.
Disassembly Part 2
Removal of pressure switch complete.
Reassembly Part 1
Prepping components for the new pressure switch.
Reassembly Part 2
New pressure switch is installed and awaiting re-connection of wiring.
Reassembly Part 3
Wiring harness assembled and wires are awaiting to be connected.
Reassembly Part 4
Wire connections on dental compressor are re-established.
Reassembly Part 5
Pressure switch cover is secured back to the unit.
Completed Midmark Apollo Compressor Repair
As well as replacing the pressure switch on this compressor, we also preformed a preventative maintenance procedure including: new oil, new air filters, and a new coalescing filter.
Air Techniques VacStar
Dual vacuum system with one failed pump that needed replacement.
Removal Part 1
Disassembly of vacuum brackets and hydromiser to remove old vacuum pump from the system.
Removal Part 2
Old pump is removed and system is prepped for the installation of the new pump.
New Electrical Wiring
Installed new electrical wiring into relay box.
New pump installed into vacuum system and tested for functionality.
Dental Chair Cuspidor
This cuspidor was leaking onto the floor when we came to the office. The drainage tubing was replaced and re-sealed to stop the leak.
Fully Functional Cuspidor
We carry the tubing and tooling necessary to replace what is needed to get this office back up and running same-day.
Broken Dental Chair Motor
Dental chair stuck in recline position.
Disassemble Dental Chair Part 1
Removal of back cover of chair to access motor wiring.
Disassemble Dental Chair Part 2
Removed the seat and backrest of the dental chair to access the motor for replacement.
Disassemble Dental Chair Part 3
Laid the dental chair on its side to access the motor compartment. Disassemble and remove the broken motor from the dental chair and replace with a functional motor.
Reassemble Dental Chair Part 1
Seat and back rest re-installed onto base and dental chair is fully functional.
Repaired and Ready for Patient
Dental chair back to full function in its upright position.
Odyssey Dental Laser
The key to turn this laser on/off was snapped off inside the laser and could not be removed. This office was told that the laser was un-repairable and needed to purchase another unit. After calling Odyssey, the new lock assembly was not available and the laser was no longer serviceable through the company.
Repair of Laser
A compatible locking mechanism was found online and we un-welded the old assembly and re-welded the new assembly onto the board.
Testing of Laser
Using the new key and new locking assembly, we were able to turn the laser and test it to make sure it is fully functional.